Article Number: 27
Created On: 2009-03-24 7:50 PM
Author: Datacolor [firstname.lastname@example.org]
URL : http://knowledgebase.datacolor.com/question.php?ID=27
This error is a Unispef error meaning black trap dianostic failed.
To determine whether is hardware or software problem try running: Terminal program located in ColorTools folder.
Choose File, Open, click on MC90.TRM
Pick : Settings, Communications. Verify settings are: 19200 baud, 1 stop bit, no parity, none flow control.
On buttons, click on: * then on Capabilities. (write down the string)
Click on *: then on Measure.
Does it have a valid measurement? Or any measurement?
You can measure black trap here with measure button and
see if it is normal, and also check white tile. If its zeroes, it’s a hardware problem.
Back to Original Question